Gas consumption: Approximately equal to preset inspiratory minute volume or equal to flow meter setting. Gas supply via high pressure inlet: From central gas supply, compressor or gas cylinder. Electronic gas supply unit, inlet pressure: 2.5-5 bar (250-500 kPa, 35-70 psi). Other gas supply inlet pressure: 2.5-7 bar (250-700 kpa, 35-100 psi). Gas supply via low pressure inlet: Vial flow meters. Power supply: (100), 110, 120, 220, 240, 50 or 60 hz. Power consumption: Approximately 40 W.
Push button: Audible signals, muted for 2 min maximum. Alarms: Gas supply alarm, apnea alarm, expired minute volume, o2 concentration, upper pressure limit, power failure. Indicators: Set minute volume alarm, set O2 alarm, set O2 alarm, pressure support, CPAP, MAN SIMV high/low rate, expired minute volume infants, patient trig.
Inspiratory pause hold, expiratory pause hold gas exchange.